Daliy weather data from mid May to late July 2011 to 2013 from Sagavanirktok Department of Transport (DOT) site (south of Toolik Field Station and Arctic LTER), in the northern foothills of the Brooks Range, Alaska. Parameters measured include: wind speed, wind directions, temperatrue, humididty, pressure and precipitation. (Rich, et al 2013).
Data set ID:
EML revision ID:
In early May through late July of 2010-2014, a weather station was placed on a tripod assembly about 200 m from the Sagavanirktok River. The site is vegetated by dwarf shrubs of Ericads and Betula. All sensors were ~ 3 meters off the ground.
Sensors are rented from Sanchez Industrial Design, Inc., and are EcoEARS data loggers with Vaisala Weather Transmitters WXT520 (Helsinki, Finland).
PRESSURE, TEMPERATURE, AND RELATIVE HUMIDITY:
Pressure: WXT520 capacitive silicon BAROCAP® sensor; range 600 ... 1100 hPa; accuracy ±0.5 hPa at 0 ... +30 °C (+32 ... +86 °F) ±1 hPa at -52 ... +60 °C (-60 ... +140 °F); output resolution 0.1 hPa, 10 Pa, 0.001 bar, 0.1 mmHg, 0.01 inHg.
Temperature: WXT520 capacitive ceramic THERMOCAP® sensor; range -52 ... +60 °C (-60 ... +140 °F); accuracy (for sensor element) at +20 °C (+68 °F) ± 0.3 °C; output resolution 0.1 °C (0.1 °F).
Humididty: WXT520 capacitive thin film polymer HUMICAP®180 sensor; range 0 ... 100 %RH; accuracy ±3 %RH at 0 ... 90 %RH; ±5 %RH at 90 ... 100 %RH; output resolution 0.1 %RH.
WXT520 uses Vaisala WINDCAP® sensor technology in wind measurement. The wind sensor has an array of three equally spaced ultrasonic transducers on a horizontal plane. Wind speed and wind directions are determined by measuring the time it takes the ultrasound to travel from each transducer to the other two.
Wind speed: range 0 ... 60 m/s; response time 0.25 s; accuracy ±3 % at 10 m/s; output resolution 0.1 m/s (km/h, mph, knots).
Wind direction: azimuth 0 ... 360°; response time 0.25 s; accuracy ±3.0°; output resolution 1°.
WXT520 uses Vaisala RAINCAP® Sensor 2-technology in precipitation measurement. The precipitation sensor comprises of a steel cover and a piezoelectrical sensor mounted on the bottom surface of the cover. The precipitation sensor detects the impact of individual raindrops. The signals from the impact are proportional to the volume of the drops.
Rainfall: cumulative accumulation after the latest auto or manual reset; collecting area 60 cm2; output resolution 0.01 mm (0.001 in); field accuracy for daily accumulation better than 5 %1, weather dependent.
Rain intensity: running one minute average in 10-second steps; range 0 ... 200 mm/h.
Reference Citations: Rich, M.E., L., Gough, N.T., Boelman. 2013. Arctic arthropod assemblages in habitats of differing shrub dominance. Ecography 36: 001-010.
Metatdata entered and data reviewed by S.Sweet, uploaded and formatted JD (Dec.2013)
Version 2: Wind speed data was a little wonky (upwards of 120+ mph). So, the folks we rent our equipment from recalculated the hourly data for us, and I finished calculating the daily averages. S.Sweet May 2014
Version 3: Added the 2014 data S.Sweet.
Download a comma delimited (csv) or Excel file (includes metadata and data sheets).
Use of the data requires acceptance of the data use policy --> Arctic LTER Data Use Policy
To cite this data set see the citation example on the LTER Network Data Portal page for this data set.